HERMLE Z233 M-2 PDF
High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||11 June 2010|
|PDF File Size:||16.64 Mb|
|ePub File Size:||4.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
The digital indications on the display will light up when the power is on. With the centrifuge lid open you can preset the operating time with the knob 3.
Hermle z m-2 Manuals
Keyboard – Starting The Centrifuge – “quick”-key 4. There is a distinction between two different kinds of errors. Preselection Of Operating Time 4.
Page 19 Error no.: Table of Contents Add to my manuals Add. Width 28 cm Depth 37 cm Height Open and reclose the lid, restart the centrifuge Error no.: Comments to this Manuals Your Name.
hermle z233 m-2 Manuals
Use only original spare parts! Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other. Page 20 Error no.: Service And Inspection Of Centrifuge 7. General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, z33, never be operated in explosive-hazardous locations.
Page 14 – Keyboard – Starting the centrifuge – “qu During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating hermoe by turning the knob 3. By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it. The green control lamps 3 show which mode is activated. Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately.
Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y. Overcurrent or undervoltage due to power supply fluctuations Action Power failure while rotor is in motion Action: Enter text from picture: Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the hwrmle Operation Operation Operation Operation Operation 4.
Don’t z2333 me this message again. Got it, continue to print. The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor. The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. Commercially available tubes or bottles made of glass or plastic.
Z M-2 Microlitre Centrifuge ( Hermle LaborTechnik GmbH ) | EVISA’s Instruments Database
On the digital display “time min-sec” the letters “ro” rotor appears. Don’t have an account? Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast. If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.
Page of 20 Go. Description – The rotor is not installed securely on the motor shaft. Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of To open the centrifuge lid.
Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation. If this is not done, they could scratch the protective coating of the rotor.
The rotor must always be balanced in this fashion prior to operation.